Maximum output power within 2W when output DC 1.2V~2V;
Maximum output power within 3W when output DC 2V~20V;
Maximum output power within 2.5W when output DC 20V~24V;
6>.Output voltage accuracy:+/-0.5%
7>.Output voltage accuracy:+/-1%
8>.Measure voltage range:DC 0~35V
9>.Measure current range:0~3A
10>.Measure power range:0~110W
11>.Measure battery capacity range:0~9999Ah
1>.The default is the Power Output Mode PER at first;
2>.Keep press button ‘ON/OFF’ to switch mode;
3>.When power is remove, module can remembers current operating mode.The default work will be the same mode when power on at next time.
5.Power Output Mode PER:
1>.Input from standard USB male socket,Micro USB female socket and ‘VIN+/VIN-’;
2>.Output from ‘OUT+’ and ‘OUT-'.The ‘COM’ terminal can not be used at PER mode.
3>.Rotate the potentiometer to adjust the output voltage;
4>.There is a symbol ‘OUT’ when work in PER mode;
5>.Terminal ‘COM’ unavailable at PER mode;
6>.Short press button ‘ON/OFF’ to turn ON or OFF output voltage.At this time, screen will display 0V.
7>.Keep press button ‘ON/OFF’ for 2 second to switch display output current or output power at the second line.
8>.Keep press button ‘ON/OFF’ for 5 second to turn ON or OFF display screen backlight.But the module can output voltage normally.
9>.Keep press button ‘ON/OFF’ for 10 second to turn switch work mode PER or VAH.
10>.The module supports over-power protection. If the rated power is exceeded, the module will automatically stop output voltage and the LCD will display ‘OPP’ and flash.
11>.The module does not support reverse connection protection, please do not short circuit and reverse.
6.Multi-function Meter Mode VAH:
1>.Module work voltage : Standard USB male socket,Micro USB female socket and ‘VIN+/VIN-’ are used to provide work voltage from module.
2>.Measuring voltage : ‘V’ and ‘COM’ terminals are used to measure Measured Device Voltage.
3>.Measuring current : ‘A’ and ‘COM’ terminals are used to measure Measured Device Current.
4>.Short press button ‘ON/OFF’ to switch display Measured Device Voltage,Current,Power,Time.And also can measure Battery capacity,Battery discharging time if connect battery.
5>.Keep press button ‘ON/OFF’ for 5 second to turn ON or OFF display screen backlight.But the module can output voltage normally.
6>.Keep press button ‘ON/OFF’ for 10 second to turn switch work mode PER or VAH.
7>.The module does not support reverse connection protection, please do not short circuit and reverse.
8>.Potentiometer is not available at VAH mode.
1>.Calibrate Voltage:Keep press button ‘ON/OFF’ before power ON and then power ON and release button.Automatically enter calibration mode after 5 second.The voltage display will flashing.Short press button to adjust and calibration voltage. Adjustment range is -0.2V~0.2V.Enter current calibration mode if keep press button for 2 second.
2>.Calibrate Current:Enter current calibration mode if keep press button for 2 second after Calibrate Voltage.The current display will flashing.Short press button to adjust and calibration voltage. Adjustment range is -0.02A~0.02A.Automatically save parameters and return to the normal display interface if keep press button for 5 second.
3>.User needs to repeat the correction multiple times. If the voltage(current) error exceeds 0.2V(0.02A).
4>.For example:Change 12V to 12.5V:
4.1>.First calibration:Change 12V to 12.2V
4.2>.Second calibration:Change 12.2V to 12.4V
4.3>.Third calibration:Change 12.4V to 12.5V
1>.Connect to power supply from input terminal;
2>.Set work mode by keep press button ‘ON/OFF’ for 10second;
3>.Connect load or Device Voltage;
4>.Rotate potentiometer to change output voltage if work in PER mode;
5>.Short press button ‘ON/OFF’ to switch parameter display;
6>.Test and use.
1>.Ordinary low power supply;
4>.Battery capacity tester;
5>.Battery load capacity tester;
1>.The module does not support reverse connection protection, please do not short circuit and reverse.
2>.Adjust the working mode before use.
3>.USB just can input DC 5V. User need input from ‘VIN+’ and ‘VIN-’ if input 3.5V-12V.
First we should say that ICStation doesn't accept any form of pay on delivery.Items used to be shipped after receiving the order info and payment. Below are the payment methods we can accept at the moment:
1) Paypal Payment
PayPal is a secure and trusted payment processing service that allows you to shop online. PayPal can be used at icstation.com to purchase items by Credit Card (Visa, MasterCard, Discover, and American Express), Debit Card , or E-check (i.e. using your regular Bank Account).
We are PayPal Verified
2) Bank Transfer / Wire Transfer / T/T
Bank Transfer / Wire Transfer / T/T payment methods are accepted for orders which the total price is up to US$500. The bank will charge about US$50 for the transfer fee if we do the payment in these ways. Feel free to contact us for bank payment details if you need pay via bank.
For other payment method, please contact us at firstname.lastname@example.org for more details.